E1QX Ethernet enabled temperature and humidity measuring device
Turning device ON
Plug one end of the Ethernet cable to the PoE enabled switch and the other end to E1QX device. The LED on the RJ45 connector of the E1QX will lit up if powered correctly.
IP configuration configuration
E1QX comes pre-configured with static IP address which is marked on the label. Use this IP address and netmask to connect to the device using standard HTTP browser such as Firefox. On the welcome page you have the option to configure IP settings of the device. Use the password to save the information. Password is statically defined on the label of the device and cannot be changed using WEB interface.
Read data from probes
Create a HTTP GET request to
http://ip_of_device/j The response will be a JSON formatted output of readings from the environmental probe. Test using curl:
curl -X GET http://192.168.10.1/j
Resetting the device to default configuration
Device has to be reset by shorting the reset pads.
- Turn off the device
- Short the reset pads on the device with a metal clip or metal tip of a pen
- While shorting the pads turn the device on
- Wait for 10 seconds and remove the shorting
- Turn the device off and on again – default IP configuration will be applied
For support please contact your distributor or manufacturer directly via www.moirelabs.com Please mention serial number and MAC address of the device in support request email.